Stainless steel or Mild steel outer with thickness 5mm or more,sheet should be painted with heat resistant and anti-corrosive paint.
Internal surface of incinerator should be line with noncracking refractorybricks made of ceramics or concrete to withstand the temperature of 1400 oC,furthermore bricks must be backed by thermal insulation and have thickness inbetween 220C250 mm.
Upto 150 kgs/hour
Upto 200 kgs
Primary Combustion Chamber
Secondary Combustion Chamber
Mix Combustion Chamber
Smoke Filter Chamber
Diesel Oil/ Natural Gas/LPG
Oil Consumption (Diesel Oil)
Gas Consumption (Natural Gas)
220 x 110 x
260 x 220 x
Oil Tank Capacity(if oil fuel)
Equipment Gross Weight
Operation Technical Specifications
Primary Chamber Temperature
Secondary Chamber Temperature
Waste Lower Calorific Power
Electrical: Single Phase Voltage 220 V, 3- Phase Voltage 400 V & Frequency 50 Hz.
Water Supply: Available pressure is 2 PSI.
Natural Gas: Available pressure 5-7 lb/in2.
Mobile main trolley of SS with cover for transportation of hospital waste fromwards to incinerator site / room. (May be provided locally)
Bin trolley Red for infectious waste Yellow for Sharps/glass and needles, Blue for non-infectious waste. Trolley Volume: 5 liter Wall thickness : 1.35 mm Trolley shouldhave flap cover (May be provided locally)
35 in each color Total Qty 105 Nos
Plastic Waste Bags Yellow, Red and blue(May be provided locally)
The flue gasses should have emission gases under limits provided by Ministry ofEnvironment, local Government and Rural development Pakistan (National Environmentaland Quality standards (NEQs) Firm will be responsible once after installation for labtesting of emission gases as per requirement of Environmental Protection Agency (EPA)Pakistan as mentioned above. Firm will be responsible for Approval of all drawings fromEPA before starting of project and certification from EPA when project is successfully done.
Automatic loading system to minimize user intervention.